Benchtop Measurement and Test
Distributed Measurement and Control
Systems Engineering Software
You can request repair, RMA, schedule calibration, or get technical support. A valid service agreement may be required.
Provides support for NI data acquisition and signal conditioning devices.
Provides support for Ethernet, GPIB, serial, USB, and other types of instruments.
Provides support for NI GPIB controllers and NI embedded controllers with GPIB ports.
Hi I'm looking for the GPIB command to get the current value from the Agilent 4294a. I'm using MATLAB over a GPIB interface. Has anyone the right command for this device?
Go to Solution.
According to the programming manual.. https://www.keysight.com/ca/en/assets/9018-01395/programming-guides/9018-01395.pdf
Reading level monitoring results p.87
To read monitoring results of oscillator (OSC) power and dc bias levels, you need to usethe following commands to turn ON the level monitor function before startingmeasurements:
• “OMON” on page 365• “BMON” on page 272
Use the following commands to read the level monitoring results:
• “OUTPVAC?” on page 394• “OUTPVACP?” on page 395• “OUTPIAC?” on page 380• “OUTPIACP?” on page 381• “OUTPDC?” on page 376• “OUTPDCP?” on page 376
Thank you so much for that!
What do you need our team of experts to assist you with?
We'll be in touch soon!