Benchtop Measurement and Test
Distributed Measurement and Control
Systems Engineering Software
You can request repair, RMA, schedule calibration, or get technical support. A valid service agreement may be required.
Provides support for NI data acquisition and signal conditioning devices.
Provides support for Ethernet, GPIB, serial, USB, and other types of instruments.
Provides support for NI GPIB controllers and NI embedded controllers with GPIB ports.
the function niDMM_Fetch sporadically returns the warning code 1073356801 (0x3FFA2001 "Over range.") with a measurement of 105 to 140 MOhm when using Auto Range. However the subsequent check with niDMM_IsOverRange returns VI_FALSE. As I understood
Can anyone explain this or is this a bug?
Thanks in advance and best regards,
Go to Solution.
So, for all those who are looking for the answer, NI R&D has confirmed this behavior as a bug (no. 2035720).
It occurs every time a resistance above 100 MOhm is measured with the PXI-4082 and Autorange enabled.
What do you need our team of experts to assist you with?
We'll be in touch soon!