메시지가 09-04-2006 11:26 AM에 _Korea에 의해 편집되었음
There may be some configuration you need to do to get the 9237 example working depending on what mode the Strain Gauge is in. If you right click on the module in the project explorer there are several configuration options available for you to chose. Make sure those are similar to what was used fo the cDAQ application. Also I noticed the example is using a DMA FIFO so make sure that you have the correct software installed on the cRIO RT controller ( i.e the current RIO drivers) and recompile the application.
Let me know if you have problems after you make sure the configuration is correct.
There is an example how to calculate strain using ni 9237 in NI labview->help--> EXAMPLEs ...I think it will be helpful to you