Aerospace, Defense, & Government
Academic & Research
Benchtop Measurement and Test
Distributed Measurement and Control
Systems Engineering Software
You can request repair, schedule calibration, or get technical support. A valid service agreement may be required.
Provides support for NI data acquisition and signal conditioning devices.
Provides support for Ethernet, GPIB, serial, USB, and other types of instruments.
Provides support for NI GPIB controllers and NI embedded controllers with GPIB ports.
I have a PXIe-5110 Scope. I am trying to simulate/build my code, however there is no option for a virtual instrument in NI MAX.
I know that the 5110 is an SDI based scope, however I'm not quite sure what that means. Will I not be able to control this scope using typical Ni Scope functions?
I am not sure where to begin.
Go to Solution.
Are you trying to simulate in LabVIEW? If so, you would open your session with the "Initialize with Options" VI instead of the "Initialize" VI, and use the options string to set up the simulation.
For a 5110, the string would be "Simulate=1,DriverSetup=Model:5110;BoardType:PXIe", and so the code would look like:
What do you need our team of experts to assist you with?
We'll be in touch soon!