Turn on suggestions

Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.

Showing results for

- Subscribe to RSS Feed
- Mark Topic as New
- Mark Topic as Read
- Float this Topic for Current User
- Bookmark
- Subscribe
- Mute
- Printer Friendly Page

Highlighted

08-10-2020 04:12 PM

Options

- Mark as New
- Bookmark
- Subscribe
- Mute
- Subscribe to RSS Feed
- Permalink
- Email to a Friend
- Report to a Moderator

Hi, I'm working with the Digitizer PXIe5160 and our goal is to capture a 100MHz signal sampled @250MS/s. However, our application is very sensitive to noise and we'd like to know how to measure the noise level of the digitizer given the fact that I'm not using the full bandwidth of the system.

I appreciate your kind help on this matter.

Cheers

Jorge Gomez

Highlighted
Options

08-12-2020 05:49 AM

- Mark as New
- Bookmark
- Subscribe
- Mute
- Subscribe to RSS Feed
- Permalink
- Email to a Friend
- Report to a Moderator

Noise (peak to peak) is a function of bandwidth, so if you reduce your bandwidth you will measure lower noise.

What information do you want to measure? (amplitude, phase/frequency)

A pure sine or some modulation? How fast change these properties (results in a bandwidth).

If you have your 5160 configured for the measurement, just terminate the input (50 ohm , or whatever your source is).

Now do some measurements and look at the noise/distribution of your result.

If you have a model of your source (signal) you can fit that to your data and/or Lock-in for periodical signals.

Greetings from Germany

Henrik

LV since v3.1

“ground” is a convenient fantasy

'˙˙˙˙uıɐƃɐ lɐıp puɐ °06 ǝuoɥd ɹnoʎ uɹnʇ ǝsɐǝld 'ʎɹɐuıƃɐɯı sı pǝlɐıp ǝʌɐɥ noʎ ɹǝqɯnu ǝɥʇ'

Henrik

LV since v3.1

“ground” is a convenient fantasy

'˙˙˙˙uıɐƃɐ lɐıp puɐ °06 ǝuoɥd ɹnoʎ uɹnʇ ǝsɐǝld 'ʎɹɐuıƃɐɯı sı pǝlɐıp ǝʌɐɥ noʎ ɹǝqɯnu ǝɥʇ'

Highlighted
Options

08-12-2020 05:48 PM

- Mark as New
- Bookmark
- Subscribe
- Mute
- Subscribe to RSS Feed
- Permalink
- Email to a Friend
- Report to a Moderator

Hi Henrik, thanks for your email. My goal is to compute the signal to noise ratio from an OFDM signal captured from the PXIe5160 digitizer. So, my first step is to calculate the noise power that the PXIe5160 gives me and then compute the power of my OFDM signal.

Just to see if I understood correctly, you're suggesting to terminate the port of the digitizer with a 50-ohm load and then do a sample capture? If so, do you suggest to have the minimum amplitude scale for the digitizer?

Cheers

Jorge Gomez