I'm using the cFP 1808 Bank along with other input modules like AI-111 and AI-110 ,I recently bought a AO-200 Module to output currents in the range from 4-20 mA ,I connected the module to the bank and updated the device from the MAX Software ,then I opened the Getting Started/Analog Output.vi from the examples in the Labview 2011 in order to test the card ,the VI returns an error with the code -33180 to me ,I don't know what's wrong ,but i tested the card with the MAX and wrote values to it successfully.
Can anybody tell me what's wrong with my VI
Solved! Go to Solution.
Post your VI so we can see what you are doing.
Did you read the error description for that error code?
I was able to reproduce your error. You have to select "All" as Field Point IO Point for this vi.