In the counter, the user has the option to select the input port through DAQmx Global Channel, start/stop and reset the count. The input i.e. square wave of 1 Hz is given by function generator for performance testing.
Counter in data acquisition devices is used to measure the number of pulses received at the given port. Port <P2.0> and <P1.1> are multiplexed as PFI0 and PFI1, respectively. This can be used as 32-bit counter input or as trigger and clock input.
Hardware and Software Requirements
NI USB-6002, LabVIEW
Steps to Implement or Execute Code
1. Connect USB-6002 through USB port.
2. The input i.e. square wave of 1 Hz is given by function generator for performance testing.
3. Download NI-DAQmx 18.0.0, Execute sample code "USB6002-counter" and run it.