Benchtop Measurement and Test
Distributed Measurement and Control
Systems Engineering Software
You can request repair, RMA, schedule calibration, or get technical support. A valid service agreement may be required.
Provides support for NI data acquisition and signal conditioning devices.
Provides support for Ethernet, GPIB, serial, USB, and other types of instruments.
Provides support for NI GPIB controllers and NI embedded controllers with GPIB ports.
About PXIe-4139 SMU:
There are two options to Switch Off Output (see attached screenshot):
Option 1 - by property node "Output Connected"
Option 2 - by "niDCPower Configure Output Enabled.vi"
According to https://knowledge.ni.com/KnowledgeArticleDetails?id=kA00Z0000019OQ0SAM&l=en-BG, Option 1 disconnect (switch off) on-board relay.
What about Option 2 ??? Does "niDCPower Configure Output Enabled.vi" disconnects on-board relay as well ???
Go to Solution.
Setting the Output Enabled property to “False” sets:
Output mode = Voltage
Voltage setpoint = 0V
Current limit = %2 of the current range.
It does not disconnect the output relay.
Thanks ChristophersonJ !
What do you need our team of experts to assist you with?
We'll be in touch soon!