It was purchased from someone who claimed it was working but I can't verify it. I don't have a second PC with long enough chassis to house this long instrument card. But the same P3 machine had been tested against several NI products in the past without problem.
Same problem exists when another PCI slot is used and I have taken a screenshot when executing the DMM Soft Front Panel, it can't proceed further and the program quit by itself. Notice that the calibration data is not visible in Max (but serial number is visible).
Greatly appreciate your assistance.
One more attempt before powering down and I managed to get past the previous error screen. Self test under utility was activated and the attached screen appeared. I believe it is a hardware fault and the card needs to be fix.