Benchtop Measurement and Test
Distributed Measurement and Control
Systems Engineering Software
Perspectives showcases how NI sees what’s next in the world of test and technology.
You can request repair, RMA, schedule calibration, or get technical support. A valid service agreement may be required.
Provides support for NI data acquisition and signal conditioning devices.
Provides support for Ethernet, GPIB, serial, USB, and other types of instruments.
Provides support for NI GPIB controllers and NI embedded controllers with GPIB ports.
Apologies in advance for the newbie questions....
I am trying to se up a test where I need a PXI-4071 to run in it's full resolution 7.5 digit DMM mode at 7Hz?, ie not waveform mode.
Using (fully updated):
- .Net 4.0 / Visual studio Express 2012
- NIDMM .net class libraries 1.0.3
- Latest DMM driver that comes with soft panel ver 14
ContinuousAcquisition.2010.sln I get it running at 100hz or more when setting resolution to 7.5 digit
SoftwareTriggeredMultipointAcquisition.2010.sln, seems to run at around 1hz, also in imediate mode. Is this the "full 7.5 digit DMM mode"? Why not 7Hz?
I am so far running the code in Visual studio Express in release mode.
Can someone please point me in the right direction, ie documentation and sample code.
Problem solved; there is an error in the demo code (ContinuousAcquisition.2010.sln)
sampleDmmSession.Configure(measurementMode, range, resolution)
sampleDmmSession.ConfigureMeasurementDigits(measurementMode, range, resolution)
Otherwise the module will run at high speed 3.5 digit resolution....
I also found another error in that code; the value of the Resource name drop-down is not read by code, i.e. if youy have more than one DMM, the application will always use the first one only.
What do you need our team of experts to assist you with?
We'll be in touch soon!