I have a NI PXI-1033 Chassis together with NI PXI 4065 DMM and NI PXI 6514 device.
The problem that I have is very annoying. When I restart my computer or install a new NI feature, the two NI PXI-6514 and PXI 4065 are no longer available as shown at the picture attached and I have to reinstall the drivers in order they to function properly. Why is this happening? Is it normal?
I will appreciate every help because it is urgent and important. Thank you!
This is definitely not normal. Let's not get into the "install a new NI feature" part... So you're saying that the devices show up in MAX correctly, you restart and then they show up crossed out? And the only way to recover is by reinstalling the drivers? Please specify driver versions as well.
I am not at my work place anymore but as far as I remember DMM 2.8, DAQ 8.8 and LabView 7.1.
Try this: open a Command Prompt and type "net start nidevldu".
Let me know what the output for this command is and whether you can use the devices after running it. Also, is there any other NI hardware in the system?
I know this topic is ancient but I'm having a similar problem with my system:
PXI-1033 (thru external PC / PCIe)
Windows 7 (32-bit)
When starting MAX it gives the MAXPXITimeout error. Then MAX crashes most of the time or when it works, none of the devices are "present" according to MAX, i.e no initialize, fail self-test, etc. I tried this command "net start nidevldu" and I get a "windows system error 5 - access denied". So I ran the command prompt as administrator and then the NI Device Loader "successfully started". Now MAX works OK and all my devices are working, until next reboot. Is this a permission issue? I tried changing the NI Device Loader service to log as an admin, but no luck. After rebooting the same issue.
Appreciate the help. Thanks,