What is the temperature dependence of the room-temperature-calibrated channel skew for the HSDIO PXI-6551? Suppose Dynamic Technology measures (after calibration or 'as left') the channel skew to be 60 ps for channel 3 at 23.2 degC. How does this value change with temperature?
There is also probably a known temporal drift. How much will this vary over a two year period?
Please advise. I'm looking into the possiblity of using the calibrated channel skew values to obtain higher accuracy.
Solved! Go to Solution.
I don't see any of this in our datasheets, I will need to look into this. Typically we have a range for how temperature affects a measurement, something like +/- X ppm/deg C.