Hi, I have the NI USB 6210 device for a low current measurement test I'm developing. I am currently trying to calibrate - what is the current in the circuit depending on the result of the USB 6210, and I noticed that when I supply low voltages then the results are not accurate...
There is this command - nidaqmx.constants.ADCTimingMode(14713) - BEST_50_HZ_REJECTION that should Improves 50 Hz noise, and I think this will solve the problem of inaccuracy at low currents.
I want to add this to the same task from which I read the value of the device.
I would appreciate it if you could help me how it can be implemented.
Or for additional ideas that can help me make the test more accurate.