Have an idea for new DAQ hardware or DAQ software features?
Browse by label or search in the Data Acquisition Idea Exchange to see if your idea has previously been submitted. If your idea exists be sure to vote for the idea by giving it kudos to indicate your approval!
If your idea has not been submitted click Post New Idea to submit a product idea. Be sure to submit a separate post for each idea.
Watch as the community gives your idea kudos and adds their input.
As NI R&D considers the idea, they will change the idea status.
Give kudos to other ideas that you would like to see implemented!
So I have a cRIO with a 9203 mA input module. I also have sensors etc that are 4-20mA. So when it came to using the scaling feature of the shared variable as below see if you can spot the bug.
So I was thinking in mA from sensor to mA input to the 9203 which is a mA module - but the RAW scale is in Amps! (Which is obvious once your colleague points it out!) Consequently I wasn't seeing any signal readings from the cRIO as 20mA << 4A.
Since there is an error if you get Full and Zero around the wrong way... can there also be an error or warning if its outside the range of the module?