Benchtop Measurement and Test
Distributed Measurement and Control
Systems Engineering Software
Perspectives showcases how NI sees what’s next in the world of test and technology.
You can request repair, RMA, schedule calibration, or get technical support. A valid service agreement may be required.
Provides support for NI data acquisition and signal conditioning devices.
Provides support for Ethernet, GPIB, serial, USB, and other types of instruments.
Provides support for NI GPIB controllers and NI embedded controllers with GPIB ports.
IMHO, it would make more sense to have the Stop Module request at the same level than Start and Sync functions.
Well I have to say that I disagree with this. The Start Module VI is not a DQMH request at all so it's separated from the other build in request like Stop Module. At first I also found this a bit confusing but it makes sense if you think about it from a DQHM point of view.
@Jens_S, the stop module is a request, yes, but the code doesn't look like all the other requests.
Moreover, I'm not sure the hierarchy needs to follow the nature of the code. For me, that's more a kind of palette organization. In this way, it makes sense for me to have the start and the stop at the same level.
Matthias Baudot | Software Architect | Founder at STUDIO BODs
Check out my LabVIEW presentations and videos!
You must be a registered user to add a comment. If you've already registered, sign in. Otherwise, register and sign in.
What do you need our team of experts to assist you with?
We'll be in touch soon!