03-12-2012 11:47 PM
I want to configure one PXI-1042Q chassis with PXI-8331 controller.I have connected 2 Digital I\O card (PXI-6515) and 1 Digital I\O card(PXI-6514) to this chassis. And I have connected this Chassis to my PC with serial cabel (NI standard Cable) on COM 1 Of my PC. But the problem is when i open NI Measurment and Automation explorer it dose not detect eithre chassi or any I\O card.
I have installed NI platform services version 4.6.1.
One obsrevation is RX led ON Chassi and TX Led ON PC dose not glow.
PLease Guide me step by step configuration.
Solved! Go to Solution.
03-14-2012 09:20 AM
The PXI-8331 isn't a serial interface even though the connector looks like it, so it shouldn't be connected to COM1. It will only work when connected to a PCI-8331 in your host PC. The cable needs to be a MXI-4 cable, such as 190957-03, not a serial cable.
If that's what you're doing, please report the LEDs on both of the cards.
03-15-2012 01:30 AM
Thank You Robert!
I Didn't want to say COM port, Its is the port of PCI 8331 Card and I have connected the system in the manner you said.
Mean time you replied this post I have Got conversation with NI engineer over Phone, He have checked system remotly and suggested to reinstall Software and NI Drivers. So I am Doing that. and Thanks for your Response.I had posted the query incorrectly. SOrry for that
06-09-2013 02:00 AM
I have connected the Pxi 1042 via Pxi8331-Pci8331,but in my PC ,MAX shows that link disconnected and chassiss status :Error 0XFFFB388C - The MXI-4 connection to this chassis timed out. Screenshot visible attachments. Or am I Inserted in the PCI slot on the motherboard (Ihave tried the both two,DELL PC ).
Please guide me how I could do ,thank you .