08-11-2011 11:36 PM
08-12-2011 01:15 AM
Ther videomaster version is 1.4.8.according to our customer spec is +- 2 degree.
but we found videomaster showd -359 degree, it's big problem for us. it should be +1 degree right ?
Thanks for your answer.
08-12-2011 01:45 AM
OK so that is quite an old version of VideoMaster and I believe that there might have been an issue in the earlier versions with just this problem. You are right that this should read +1 degree. I would recommend that you contact your local NI office and ask about updating your version to the latest version. I have made this measurement with later versions than this and have not seen this issue.
Hope this helps,
08-30-2011 10:57 AM
Hi Jeff2 Lee,
You would need to update to the latest version which would be V3.1. I think the issue was fixed in either V3.0.1 or V3.0.2. There may also be a cost involved in this update but you would need to check with your local NI guys.