04-29-2010 09:05 AM - edited 04-29-2010 09:10 AM
Hello to everybody!
I've a question about using a DMM (PXI-4065) with VStudio 2008 and Measurement Studio 2009.
Because DMM are not still supported in .Net, I downloaded niDMM.cs wrapper 3.0 from NI web site.
I also downloaded C# Measure DC Volts example to compare it with my application.
Both, example and my application, fail trowing the exception [-1074118656] "Device was not recognized. The device is not supported with this driver or version." when constructor class is called!
dmm = new InstrumentDriverInterop.Ivi.nidmm("PXI3::12::INSTR", true, true);
I suppose it's depending on PXI device!
I also tried to use a VISA alias like "pippo" and then I tried to insert an unexistent device name. In any case the same exception has trown.
Solved! Go to Solution.
04-30-2010 04:45 AM
Maybe you areusing a wrong itrument descriptor, please try the instructions in the following document:
Let me know if it could help.
04-30-2010 05:24 AM
Thank you Matteo
That was the solution!
Now I'm using the name in "NI-DAQmx Devices" MAX tree [PXI1Slot5] instead of the PXI MAX tree [PXI3::12::INSTR].