02-29-2012 02:50 PM
I have connected GPIB to the EG&G lock in amplifire. Then I scan the instrument using Measurement and Automation analyzer to check the instrument number? But I get a different number everytime i scan the instrument. For your better understanding to my question, I have attached a file and rounded the identification numer, which chaged everytime i scan. Can you please give me a solution to this problem?
02-29-2012 02:53 PM
03-06-2012 04:08 PM
Thanks 4 reply. I am able to identify the GPIB address, but it is not showing instrument name. It is showing like +00000, +00000. This number keeps on changing. I also tried to make a programme and start running the programme. However, the programme just stuck after initialization. Plz help me to get rid of this problem
Thanks in advance
03-06-2012 04:56 PM
The fact that MAX does not display an instrument name is not important. As Albert said, the instrument does not support the *IDN? query. You simply ignore what is there.
Unless you attach what you wrote, no one can help you fix a problem. Obviously, any code that contains a *IDN? query is going to be invalid.
03-08-2012 01:21 PM
Thanks for reply. If you see the attached file, u can know that the instrument is not responding even after giving a correct gpib address.