Have an idea for new DAQ hardware or DAQ software features?
Browse by label or search in the Data Acquisition Idea Exchange to see if your idea has previously been submitted. If your idea exists be sure to vote for the idea by giving it kudos to indicate your approval!
If your idea has not been submitted click Post New Idea to submit a product idea. Be sure to submit a separate post for each idea.
Watch as the community gives your idea kudos and adds their input.
As NI R&D considers the idea, they will change the idea status.
Give kudos to other ideas that you would like to see implemented!
PCI Express bus became more and more popular. PCIe has a new type of interrupts - MSI/MSI-X. Today VISA driver is able to work only with old style interrupts (Legacy).
Let me explain the main difference:
Legacy interrupt mask intA/B/C/D signals (these signals was in PCI bus, but not exist in PCIe bus). These signals (for PCIe actually packets with Message setA/B/C/D) are shared between all PCIe devices. So VISA driver spend a lot of time when it looks who produced this interrupt.
MSI interrupt is actually a Memory Write packet to preallocated address. In this case VISA should already know which device produced this interrupt. Also MSI interrupt can contain different interrupt vectors inside of Memory Write packet. So it would be very helpful to get access to vector value too.
Requesting MSI/MSI-X interrupts support to VISA driver.
Just ran into a situation where I need to stream a lot of data to TDMS. The only problem is that I need to store additional metadata with the channels. I could go through all of the generated TDMS files and insert them after the fact, but this is kind of tedius. I propose a way to add metadata to the channel. My first thought was to use a variant input on the Create DAQmx Channel, but some of the polymorphics already have really fully connector panes. So I am now thinking to just add a property to the Channel Property Node that is just a variant. When logging to TMDS, the variant attributes can be put in the metadata of the channel. Do something similar for the group so that we can have additional group metadata.
Metadata that I'm currently thinking about could include sensor serial number and calibration data. I'm sure there is plenty of other information we would like to store with the TDMS file.
By default, DAQmx terminal constants/controls only show a subset of what is really available. To see everything, you have to right-click the terminal and select "I/O Name Filtering", then check "Include Advanced Terminals":
I guess this is intended to prevent new users from being overwhelmed. However, what is really does is create a hurdle that prevents them from configuring their device in a more "advanced" manner since they have no idea that the name filtering box exists.
I am putting "advanced" in quotes because I find the distinction very much arbitrary.
As a more experienced DAQmx user, I change the I/O name filtering literally every time I put down a terminal without thinking about it (who can keep track of which subset of DAQmx applications are considered "advanced"). The worst part about this is trying to explain how to do something to newer users and having to tell them to change the I/O name filtering every single time (or if you don't, you'll almost certainly get a response back like this).
Why not make the so-called "advanced" terminals show in the drop-down list by default?
It would be great if the full DAQmx library supported all NI data acquisition products on Windows, Mac OS X and Linux. The situation right now is too much of a hodge-podge of diverse drivers with too many limitations. There's an old, full DAQmx library that supports older devices on older Linux systems, but it doesn't look like it's been updated for years. DAQmx Base is available for more current Linux and Mac OS systems, but doesn't support all NI devices (especially newer products). DAQmx Base is also quite limited, and can't do a number of things the full DAQmx library can. It's also fairly bloated and slow compared to DAQmx. While I got my own application working under both Linux and Windows, there's a number of things about the Linux version that just aren't as nice as the Windows version right now. I've seen complaints in the forums from others who have abandoned their efforts to port their applications from Windows to Mac OS or Linux because they don't see DAQmx Base as solid or "commercial-grade" enough.
I'd really like to be able to develop my application and be able to easily port it to any current Windows, Mac or Linux system, and have it support any current NI multi-function DAQ device, with a fast, capable and consistent C/C++ API.
It is a frequent requirement to make measurements on production lines. Position on these is often tracked with Rotary Encoders https://en.wikipedia.org/wiki/Rotary_encoder . Many NI devices can accept the quadrature pulse train from such a device, and correctly produce a current position count. The information in the 2 phase pulse train allows the counter to correctly track foward and reverse motion.
What would be very useful would be a callback in NI-DaqMX that is called after every n pulses, ideally with a flag to indicate whether the counter is higher or lower than the previous value, i.e. the direction.
Since the drivers support the basic communication for the GPIB-USB-HS+ devices, the code should be ported to support the analyzer functionality as well. Create the driver API to support analyzer and then make that available to LV. Create cross platform LV routines for the analyzer functionality so it is available on all platforms.
When using TDMS on cRIO systems, there are a couple of considerations that doesn't normally play in too much when storing data as TDMS files and they are:
* The current version of the file system used on cRIO controllers degrades significantly in performance if -any- folder on the cRIO contains more than ~100 files. (work-around> more elaborate folder structures, but a lot of this structuring would be only to work around this shortcoming of the (old) version of the file system
* The drives are SSD with limited life-length and wear-leveling etc. Writing and re-writing these index files add un-necessary overhead and wear on the disks
* They use up space which is (very) limited on some cRIO's (even if not much). (people may be quick to point out that you can add a thumb-drive, but down-sides to that is the thumbdrives (as far as I know) needs to be FAT. Compared to storing on the cRIO file system which is atomic and fail-safe where you pretty much don't have to worry about sudden power outages and interruptions mid-write.. on a thumb drive you would have all these issues that could worst case corrupt your whole thumb drive.)
I propose to add a boolean (default to false) on the TDMS Open called "supress writing tdms index to disk" or some smart name along those lines. What this would do is still allow for the tmds index to be created, but it will remain in memory only and never be written to disk. When the TDMS Close is called, the memory is released and the tdms file is written to disk without the index file. If the same file is opened again, extra time would be needed since the index file would be re-created (again in memory only if boolean indicates this), but I think for the most part this overhead would be more than acceptable.
I'm not sure how "simple" modifying the TDMS open and close functions would be, but I do know that there are many cases where this flag would make sense.
Every time I have to work with a NI daq device the first thing i need to know is what pins can or cant do something.
Currently this involves looking through something like 7 diffrent documents to find little bits of information and bringing them back to your applicaiton.
A block diagram could easily be a refrence point for the rest of the documentation (you want to know about pin IO for your device look at this document)
Plus a good block diagram can tell you what you need to know quickly, and clearly. A picture is worth 1000 words?
Some might find the current documentation adiquite, but personally i would really like to have a block diagram that represents the internals and capiblities of the pins and device in general. Most Microcontrollers have this and it is an extremly useful tool. So why not have one for the Daq devices as well?
There isn't anything in the event data node to tell you which line triggered the interrupt. I'm proposing we add something in the event data node for this event (like a bit field or a reference to the channel) so the programmer would know which line fired the event.
The workaround is you do a DAQmx read at this point and you mask the data vs previous data.. but I would prefer not to do this.
I bought a NI USB-6251 BNC but the support explained me that it would have no Linux support out of the box. I will have to find out how to use it on Linux systems myself now (perhaps with help of the forum). It would be a nice feature, if it would ship with Linux support.
We need a way to query an output task to determine its most recently output value. Or alternately, a general ability to read back data from an output task's buffer.
This one's been discussed lots of times over the years in the forums but I didn't see a related Idea Exchange entry. Most of the discussion I've seen has related to AO but I see no reason not to support this feature for DO as well.
There are many apps where normal behavior is to generate an AO waveform for a long period of time. Some apps can be interrupted unexpectedly by users or process limit monitoring or safety range checking, etc. When this happens, the output task will be in a more-or-less random phase of its waveform. The problem is: how do we *gently* guide that waveform back to a safe default value like 0.0 V? A pure step function is often not desirable. We'd like to know where the waveform left off so we can generate a rampdown to 0. In some apps, the waveform shape isn't directly defined or known by the data acq code. So how can we ramp down to 0 if we don't know where to start from? This is just one example of the many cases where it'd be very valuable to be able to determine the most recently updated output value.
Create a DAQmx property that will report back the current output value(s). I don't know if/how this fits the architecture of the driver and various hw boards. If it can be done, I'd ideally want to take an instantaneous snapshot of whatever value(s) is currently held in the DAC. It would be good to be able to polymorph this function to respond to either an active task or a channel list.
Approach 2 (active buffered tasks only):
We can currently query the property TotalSampPerChanGenerated as long as the task is still active. But we can't query the task to read back the values stored in the buffer in order to figure out where that last sample put us. It could be handy to be able to query/read the *output* buffer in a way analogous to what we can specify for input buffers. I could picture asking to DAQmx Read 1 sample from the output buffer after setting RelativeTo = MostRecentSample , Offset = 0 or -1 (haven't thought through which is the more appropriate choice). In general, why *not* offer the ability to read back data from our task's output buffers?
In my post on the LabVIEW board I asked if it was possible to have control over the DIO of a simualted DAQ device. Unfortunately it seems this feature is not available. Once MAX is closed the DIOs run through their own sequences.
If there was a non-blocking way to control a simulated DAQ device through MAX it would permit much simpler prototyping of systems before they need to be deployed to hardware. For example if you want to see how a program responds to a value change simply enter it in the non-blocking MAX UI. Or as in my original case can make an executable useable even if you don't have all the necessary hardware.
I think this feature should only be available for simulated devices.
Any series card should have a feature listing different parameters like voltage, temperature etc it supports(May be a property node should be used). so that user can configure the required parameter among the supported.
Ex: SCXI -1520 module can be configured as Strain, Pressure or voltage but this information will be known only by seeing its manual or when a task is created in MAX. But in LabVIEW Software i cant get this information directly. Because it allows me to configure 1520 as temperature also and we will come to known that 1520 module doesn't support for temperature parameters only when once tried to acquire.
So what you people think about you.Share your ideas on this please.
Currently when streaming analog or digital samples to DAQ board, output stays at the level of last sample received when buffer underflow occurs. This behavior can be observed on USB X Series Multifunction DAQ boards. I have USB-6363 model. The exact mode is hardware-timed, buffered, continuous, and non-regenerating. The buffer underflow error code is -200290 “The generation has stopped to prevent the regeneration of old samples. Your application was unable to write samples to the background buffer fast enough to prevent old samples from being regenerated.”
I would like to have an option to configure DAQ hardware to immediately set voltage on analog and digital outputs to a predefined state if the buffer underrun occurs. Also, I would like to have an option to immediately set one of PFI pins on buffer underrun.
I believe this could be accomplished by modifying X series firmware and providing configuration of this feature in the DAQmx API. If no more samples are available in the buffer the DAQ board should immediately write predefined digital states / analog levels to outputs and indicate buffer underrun state on PFI line. Then it should report error to PC.
Doing this in firmware has certain advantages:
It can be done quickly (possibly within the time of the next missing sample – at 2Ms/s that’s 0.5us).
Handles all situations (software lockups, excessive CPU loading by other processes, loss of communication do to bus traffic, interface disconnection…)
It does not require any additional hardware (to turn off outputs externally).
Buffer underrun indication on PFI line could provide additional safety measure (it could be used for example to immediately disable external power amplifier connected to DAQ AO).
Doing this using other methods is just too slow, does not handle all situations, or requires additional external circuitry.
Setting outputs from software, once error occurs, is slow (~25ms / time of 50000 samples at 2MS/s) and does not handle physical disconnection of the interface. Analog output does eventually go to 0 V on USB-6363 when USB cable is disconnected, but it takes about half a second.
Using watchdog timer would also be too slow. The timer can be set to quite a short time, but form the software, I would not be able to reset it faster than every 10ms. It also would require switching off analog channels externally with additional circuitry, because watchdog timer is not available for analog channels.
The only viable solution right now is to route task sample clock to PFI and detect when it stops toggling. It actually does stop after last sample is programmed. Once that occurs, outputs can be switched off externally. This requires a whole lot of external circuitry and major development time. If you need reaction time to be within time of one or two samples, pulse detector needs to be customized for every possible sampling rate you might what to use. To make this work right for analog output, it would take RISC microcontroller and analog electronic switches. If you wanted to use external trigger to start the waveform, microcontroller would have to turn on the analog switch, look for beginning of waveform sample clock, record initial clock interval as reference, and finally turn off the switch if no pulse is received within reference time.
I’m actually quite impressed how well USB-6363 handles streaming to outputs. This allows me to output waveforms with complexity that regular arbitrary generators with fixed memory and sequencing simply cannot handle. The buffer underflow even at the highest sampling rate is quite rare. However, to make my system robust and safe, I need fast, simple, and reliable method of quickly shutting down the outputs that only hardware/firmware solution can provide.
The title pretty much says it all. I would like the ability to either configure a full hardware compliment as simulated devices then switch them over to real devices when the hardware arrives or go from real devices to simulated devices without the need to add new, discrete simulated devices to MAX.
This would make for much easier offline development and ultimate deployment to real hardware.
When it comes to documentation of an measurement, you need to report ALL settings of a device that effects that measurement.
From a core memory dump written as a hex string to a XML document.... anything that shows up a difference in the settings that affect the measurement would be fine for documentation.
Something like a big property node readout followed by a format into string .... but make sure not to miss a property.... and a bit more complicated when it comes to signal routing....
A measurement that isn't sufficiently documented is all for naught.
Just think of a nasty auditor
It's so easy to make measurements with LabVIEW, please make it easy and consistent to document it.
A quick measurement setup with the DAQ-assistant/Express fills Gigabytes but after a certain time they are useless because nobody knows how they where taken. A simple checkbox could add all this information in the variant of the waveform. (or TDMS or ...) even if the operator don't have a clue of all the settings that affect his measurements.